<--2016-04-13 09:05:00 |
Mayberry Wetland: 2016-04-28 09:25:00
Summary: Standard data collection, deployed new conductivity sensor, serviced miniDots
Author: Elke Eichelmann
Others: Dennis Baldocchi, Kyle Hemes
|Arrived at 9:25. Sunny and windy day, and very clear, we were able to see the Sierra from the bridge. Vegetation is visibly shorter than previously, wind seems to have blown over the old vegetation and the new stuff coming through is not that tall yet. There is lots of detritus and the wetland is just starting to green up.|
Filled LI7700 wash water reservoir, was almost empty.
Reconnected new conductivity sensor CS547A sn 5688. Seems to be working. We did not know where the original conductivity sensor was placed in the water, so we just dropped it in the water by the tower. Can be moved if necessary.
Collected USB GHG data, collected met data from CR1000, collected camera card pictures.
All data looks good except Ts16 is reading INF.
Serviced miniDots and downloaded the data. Replaced the missing screen on one of the water CO2 sensors.
SS 83.331 -> 85.639 after cleaning
RSSI: 32.30 -> 71.42 after cleaning
Solar panel: Amp draw 1.5A; Volt 13.9
Bring some string next time to attach bolt for camera arm to scaffolding to prevent it from dropping in the water.
Left at 10:15
0 photos found
Graphs display: 7 days |
14 days |
9 graphs found
6 MB_flux graphs found
No data found for MB_met.
14 MB_tule graphs found
Can't check MBWPEE_mixed
2 MBWPEE_mixed graphs found
1 MB_cam graphs found
Can't check MBTAEEWPTW_cam
1 MBTAEEWPTW_cam graphs found
Can't check SIMBTWEE_met
1 SIMBTWEE_met graphs found
2 MB_minidot graphs found
Can't check EEMBTW_mixed
1 EEMBTW_mixed graphs found