<--2017-08-30 09:50:58 |
East End: 2017-09-13 10:05:53
Summary: Main tower: swapped 7500, removed conductivity and HOBO/miniDOT sensors
Boardwalk: sonic spikes faced downward
Closed canopy: deployed HOBO/miniDOT sensor
Open water: water level lower and less plant cover, sonic spikes faced downward
Author: Daphne Szutu
Others: Elke Eichelmann, Sam Chamberlain
It was sunny with some clouds and a light breeze. There was a small tent set up by the side of the road, between West Pond and the Closed Canopy Temporary Tower. We could tell there was a person inside, and there were two small, square electronic devices in front of the tent. We didn't see a vehicle.
Later, we drove by Juan. He explained that the person shouldn't be there, and they would take care of it. We should check in with him or Bryan before our next trip to the wetlands.
Temporary Tower 2 – Closed Canopy
We arrived at 9:55. It was sunny with a light breeze. The GPS location is: 38.106032, -121.641761.
There were no wasps in the charge controller box. The wash reservoir was still full, and the box was full of wasps so we left the box open.
Sam cleaned the EC sensors. I downloaded met and USB GHG data.
SS 90.3 --> 93 after cleaning
RSSI: 47.5 --> 73
We left at 10:05.
Sam came back to deploy the HOBO and miniDOT sensors around 11:45.
We arrived at 10:05. The cattails are still mostly green but browning at the tips.
The conductivity sensor values are flat; it is underwater and cleaned two weeks ago. I pulled the sensor sn 6494 and module sn 3585 and will test it in the lab.
I unscrewed and re-screwed the 7550 temperature sensor.
Elke refilled the wash water; it was mostly full. She cleaned the EC and radiation sensors.
We swapped the LI-7500. sn 75H-2150 came down, sn 0041 going up. I downloaded the complete LI-7550 configuration file, "sn2150_0397_EE-all_20170913.l7x". I uploaded a new configuration file with zero/span and calibration coefficient values for sn 0041 and updated the pressure coefficients. The LI-7550 software is version 8.0.0, and I did not update the software.
I downloaded met, camera, and USB GHG data. I also downloaded the miniDOT and HOBO data. I re-launched the HOBO sensor to clear the data.
CO2 391.5 --> 405.9 after swap
H2O 832 --> 766.6 after swap
T 23.0 --> 22.1 after swap
P 101.0 --> 101.6 after swap
SS 89.2 --> 97 after swap
RSSI: 37.2 --> 81.8 after cleaning
Sam collected pore water samples while Elke and I went to the Boardwalk tower. He came back later in the evening to finish sampling [Sam - more details].
We left at 11:00.
Temporary Tower 3 – Boardwalk
We arrived at 10:45. The reservoir was 1/2 full; Elke refilled it. There were some wasps so we left the box open. There were also wasps in the solar charge controller box. We put in a few mothballs and closed the box.
Elke cleaned the EC sensors and righted the sonic bird spikes. The spikes were hanging downward, but not quite in the path of the sonic. The sonic top is bloody.
I downloaded met and CR1000 GHG data.
RSSI: 35 -> 76 after cleaning
We left at 11:00.
Temporary Tower 1 – Open Water
We arrived at 11:10. It was sunny with a light wind. The water level was lower than two weeks ago and less of the water is covered with the moss-like aquatic plant.
I collected the met data and swapped the met battery. The wash reservoir was mostly full. Elke cleaned the EC sensors but could not reach the top 7700 mirror. Some sonic spikes were knocked over, but not in path's way.
SS 105 --> 109 after cleaning
RSSI: 52 --> 68
We left at 11:40.
0 photos found
Graphs display: 7 days |
14 days |
9 sets found
10 EE_met graphs found
Can't check EE_mixed
1 EE_mixed graphs found
8 EE_flux graphs found
Can't check MBWPEE_mixed
2 MBWPEE_mixed graphs found
1 EE_cam graphs found
Can't check MBTAEEWPTW_cam
1 MBTAEEWPTW_cam graphs found
Can't check SIMBTWEE_met
1 SIMBTWEE_met graphs found
Can't check EEMBTW_mixed
1 EEMBTW_mixed graphs found
Can't check EE_Temp
5 EE_Temp graphs found